The Synergy HT Multi-detection reader is a robotic-compatible microplate .. and J. Sambrook () Molecular Cloning A Laboratory Manual, Cold Spring. View and Download BioTek Synergy HT operator’s manual online. Multi-Mode Microplate Reader. Synergy HT Laboratory Equipment pdf manual download. Manuals and User Guides for BioTek Synergy HT. We have 2 BioTek Synergy HT manuals available for free PDF download: Operator’s Manual.
|Published (Last):||14 November 2014|
|PDF File Size:||2.77 Mb|
|ePub File Size:||6.12 Mb|
|Price:||Free* [*Free Regsitration Required]|
Either cable can be plugged into either socket. Wipe with lens-cleaning tissue and set aside.
BioTek Synergy HT Manuals
Cleaning The Optical Probes Perform these steps to prepare the dispense module for shipment: Chapters Table Of Contents Warranty and Product Registration Take a moment to review the Warranty information that shipped with your product. Performance Check After reassembling the instrument, perform the following to verify that the instrument is functioning properly: Manhal 45 Hold the syringe from rotating while tightening the thumbscrew.
If you have not already done so: Creating Protocols And Bjotek When the read is complete, measurement values will appear in Gen5. Computer Control The Synergy HT is completely computer-controlled; it is not equipped with a keypad or its own user interface.
Le present appareil numerique n’emet pas de bruits radioelectriques depassant les limites applicables aux appareils numerique de la Class A prescrites dans le Reglement sur le brouillage radioelectrique edicte par le ministere des Communications du Canada. Home Sensor Verification Errors Warranty And Product Registration Turn on the instrument.
Wipe all sides of the plate carrier. Chapter 2 Installation This chapter includes instructions for unpacking and setting up the Synergy HT and, if applicable, the external dispense module. We recommend that you perform these tests before first use e. The slowest of the three available modes. Run the Absorbance Plate Test Hardware And Environmental Internal Self-test Errors 0d Results And Troubleshooting Tips Install the Fluorescence Lamp Assembl If you suspect that a Soak the probe in alcohol for one minute maximum.
BioTek Synergy HT Operator’s Manual
This solution can also be made by diluting the BioTek wetting agent Install Gen5 Software Installation Product Registration Please register your product with BioTek to ensure that you sybergy important information and updates about the products you have purchased. Robotic Interface This section shows the location of the microplate carrier in reference to the exterior surfaces of the Synergy HT and the mounting holes on the bottom.
Gen5 also manial certain configuration items that must be set before using the dispense module. Page Appendix C: Page of Go.
Biotek Synergy HT Manuals
This error is caused when the Order-sorting Filter Wheel cannot rotate to the home sensor position. Hold the syringe from rotating while tightening the thumbscrew.
Installation BioTek Instruments, Inc. COM port to which the reader is connected. Contact BioTek Customer Care with any questions. For Use with Liquid Tests see Chapter 5. synrgy
Enter text from picture: Internal plate shaking is also supported. Use two people when lifting and carrying the instrument. Page 54 Perform these steps to prepare the dispense module for shipment: Test The Injector System Chapter 2: Read the following sections to become familiar with these functions and configuration items.
This error indicates that the combination of assay parameters results in more kinetic reads than supported by the software. Don’t have an account?
BIOTEK SYNERGY HT OPERATOR’S MANUAL Pdf Download.
Electromagnetic Interference And Sjnergy Page Clean the Internal Components Lift the ground wire and move it off to the side. Remove both of them and set them aside.
Recommendations For Optimum Performance Gen5 protocol in advance as well, to ensure that the intended reading parameters are used and to avoid any last-minute corrections.
Canadian Department of Communications Class A Installation Dispense module Top synefgy cable end cap Dispense module cover Inlet tubes 2 Syringes 2 Outlet tubes 4 Bottom foam end cap with cutouts Accessories Figure 9: